The vast majority of imaging experiments measure fluorescence intensity, but there is another key information-rich property of fluorescence—its lifetime. Lifetime information comes “for free” with every fluorescence experiment. It can provide valuable new insights, yet in most cases it is never collected due to technical limitations. Until recently, lifetime imaging was considered too slow, complex and expensive for everyday microscopy applications, especially those that involve live cell imaging. But all that is changing fast. With today’s advanced technologies, lifetime imaging is becoming faster and easier to use than ever before. In this review, we explore what lif time imaging is and the advantages it can bring to your research across a wide range of application areas.
What is fluorescence lifetime?
Ever since the invention of the microscope, biologists have been exploiting the properties of light to see inside cells. Fluorescent molecules have become essential tools for cell biologists. They are widely used in microscopy to visualize many di ferent structures, targets and dynamic processes within cells, tissues and even whole organisms. Over the past century, researchers have developed a diverse toolbox of fluorescent probes, with colors (emission wavelengths) that span the
spectrum from UV to far red.
But there is more to fluorescence than color. Each species of fluorescent molecule also has a characteristic lifetime that reflects how long the fluorophore spends in the excited state before emitting a photon (Figure 2). For a given fluorophore population, photons will be detected at a range of times after a pulse of excitation. For each fluorophore species, the distribution of these observed lifetimes follows an exponential decay function, from which a fluorescence decay constant τ can be calculated.
In fluorescence lifetime imaging (FLIM), τ is determined at every pixel, and then translated into a spatial map. This process results in a lifetime image that looks just like a conventional intensity image. However, instead of representing fluorescence intensity, each pixel represents the value of τ at that location of the sample.
Why is fluorescence lifetime useful?
When it comes to understanding what is happening in your specimen, fluorescence intensity measurements tell you only half the story. Lifetime analysis provides a wealth of additional information to help improve the quality of routine confocal experiments, give you more freedom of choice in probe selection, and enable novel applications.
The lifetime parameter is particularly useful for distinguishing different fluorescent species within the same sample. Even if two fluorescent dyes have the same color (i.e. closely overlapping emission spectra), it may be possible to distinguish them on the basis of their lifetimes. Moreover, you can make use of lifetime information from natural autofluorescence within your samples to enhance image contrast
and run label-free experiments.
Unlike fluorescence intensity, lifetime is concentration-independent. This can be a big benefit in cases where probe concentration is unpredictable or variable across the sample. At the same time, lifetime probes can be exquisitely sensitive to local environmental conditions within your sample, enabling you to monitor dynamic changes in intracellular temperature, pH, ion concentration, polarity and many other parameters that would be difficult or impossible to follow by conventional techniques.
Best of all, the lifetime parameter is an intrinsic characteristic of flfluorescent molecules, so lifetime data is available “for free” with every fluorescence experiment. You just need to collect it!
Four ways FLIM can improve your current experiments
1. Reduce the chance of artifacts
The intensity of a fluorophore varies with concentration, light source strength, and many other variables that may have nothing to do with the phenomenon you want to study. A major advantage of fluorescence lifetime is that it is independent of fluorophore concentration and other factors that limit the utility of intensitybased measurements.
A fluorophore will always exhibit the same lifetime in a particular microenvironment, regardless of its concentration or the method of measurement.
Thus, by measuring fluorescence lifetime, you can reduce artifacts caused by photobleaching, non-uniform illumination, variable instrument settings (e.g. light path, laser intensity, exposure time, detector gain), varying probe depth within thick samples, etc.
2. Differentiate true signal from unwanted autofluorescence
Virtually all cells contain structural components and metabolites that fluoresce when excited at wavelengths across the UV/visible spectrum. This naturally occurring “autofluorescence” often overlaps with the emission of common fluorescent probes, but typically it has a very different lifetime “signature”. In such cases, lifetime imaging can provide a means of differentiating the true signal from the autofluorescence.
3. Improve contrast
So-called “unwanted” autofluorescence can contain valuable information. Rather than discard or ignore it, you can use the lifetime data from autofluorescence to improve contrast and reveal greater detail in your sample. For instance, anatomical features in a histological cross-section from a nematode can be difficult to distinguish from intensity information alone, but appear much more
distinct with lifetime imaging (Figure 3).
4. Distinguish more fluorophores with confidence
A key element to success in multi-probe imaging experiments is to choose a color palette with minimal spectral overlap. Unfortunately, many popular fluorescent dyes have overlapping spectra. If your confocal microscope has lifetime imaging capability, you can use it to better distinguish the different probes in your sample. This not only improves the quality of your results, it also gives you more options when it comes to dye selection.
Lifetime imaging adds new dimensions to your research
In addition to enhancing everyday confocal experiments, lifetime imaging provides a means of visualizing and quantifying aspects of physiology and function that you would struggle to study using intensity-based or ratiometric measurements. Here are 3 popular application areas where lifetime imaging can help you ask new types of experimental questions and gain novel insights.
1. Analyze metabolism with label-free experiments
Label-free analysis of autofluorescence for functional imaging - particularly to study metabolism - is a rapidly growing research area. Professor Ammasi Periasamy, Director of the Keck Center for Cellular Imaging at the University of Virginia, has done significant research into the use of FLIM in combination with 2-photon excitation for functional imaging. He applies label-free lifetime imaging in his current research to measure metabolism and mitochondrial dysfunction in living cells, tissues and animals. Coenzymes central to metabolism such as FA D and NADH provide natural autofluorescence (Figure 4). “You don’t have to label it,” says Prof. Periasamy, “so this type of technique is useful for translational research.” Importantly, he points out, lifetime discriminates the bound and free states of these coenzymes. “With intensity techniques it is impossible to do that.”
2. Study molecular interactions with FRET-FLIM
Interactions between proteins, DNA, RNA and other molecules underlie much of biology. A phenomenon known as Förster resonance energy transfer (FRET) can be exploited to detect molecular interactions and measure the distance between molecules at very close range - over distances less than ~10 nm.
Beyond its utility for measuring intermolecular distances, the principle of energy transfer between FRET donor and acceptor pairs has also enabled development of sensitive biosensors. Many of today’s modern FRET biosensors are based on genetically encoded proteins  engineered to undergo a conformational change that
brings linked donor and acceptor domains closer together upon ligand binding, so that a FRET signal is generated.
Genetically encoded FRET biosensors are particularly useful for monitoring dynamic signal transduction events in live cells. For instance, the ERK signaling pathway is a key driver of malignant cell growth in cancer. A FRET biosensor for ERK has been used to monitor cancer-associated changes in ERK signaling in live patientderived organoids . Lifetime analysis highlights enhanced ERK activity in organoids from colorectal cancer patients (Figure 5). High-speed FLIM recording was crucial for this demanding 3D imaging study.
3. Explore microenvironments with environmentally sensitive probes
An especially useful property of fluorescence lifetime is its potential sensitivity to the local microenvironment. This phenomenon is possible because the surrounding environment can affect the available pathways by which excited electrons return to the ground state.
If environmental factors increase or decrease non-radiative pathways (those that do not result in emission of a photon), then the lifetime of the fluorophore will be shortened or lengthened respectively.
A wide variety of environmental biosensors have been engineered to exploit this property - many of them based on FRET. In many cases, lifetime imaging reliably detects environmental changes that are otherwise extremely difficult to measure in the crowded intracellular space. Specific environmental sensors are available to monitor temperature, ion concentration (Ca2+, Na+, etc.), pH, polarity, viscosity, second messengers such as cAMP, and many other parameters and small molecule analytes.
A particularly challenging application where lifetime-based biosensors are shedding new light is in studying the temperature-dependence of biological activities within sub-cellular compartments such as mitochondria and the nucleus, where vital cell functions occur. Traditional thermometry techniques cannot be applied at the single-cell level, where regions of interest have dimensions ranging from 1μm to ~100μm. To address this problem, researchers have developed molecular thermosensors such as the fluorescence polymeric thermometer (FPT), a concentration-independent cell permeable polymer that exhibits an increase in fluorescence lifetime with increasing temperature .
Dr. Kohke Okabe and colleagues combine FPT with time-correlated single photon counting (TCSPC) - one of the most accurate FLIM detection approaches - to determine whether highly localized temperature variations affect cellular responses such as stress granule formation and brain edema. “FLIM has been indispensable
to do our work in a quantitative manner,” says Okabe.
Summary and Conclusion
Integrating FLIM into everyday research Lifetime imaging is an invaluable tool that can immediately enhance the quality of your confocal experiments. Not only does it allow you to take advantage of natural cellular autofluorescence, it also gives you much more flexibility in probe selection and multiplexing experiments. The emergence of novel FRET-FLIM and environmental probes opens up a world of additional possibilities for cell research and quantitative imaging, enabling scientists to ask new questions and quantify aspects of cell structure and physiology that could not previously be measured. As lifetime imaging technology continues to evolve, previous limitations such as photobleaching and slow data acquisition speeds are becoming less of an issue. Rapid video rate FLIM solutions such as FALCON technology  are enabling more live-cell kinetic applications. In the future, we can expect to see more holistic confocal platforms that seamlessly integrate FLIM into everyday imaging workflows. This will be facilitated by more user-friendly software and turnkey approaches to system operation, experimental set-up, data analysis and results interpretation. Periasamy predicts a bright future for FLIM technology in biological research: “This is a dynamic technique that will surely continue to advance. It will be available in all biology labs and microscopy facilities - just like confocal [is today].”
Download Review Article "Exploring Hidden Dimensions of Cell and Tissue Structure and Physiology with FLIM" (pdf)
- Greenwald EC, Mehta S and Zhang J. Genetically encoded fluorescent biosensors illuminate the spatiotemporal regulation of signaling networks. Chem Rev 118, 11707-11794 (2018).
- Alvarez LAJ et al. Application Note: SP8 FALCON : a novel concept in fluorescence lifetime imaging enabling video-rate confocal FLIM. Nat Meth (2019)
- Okabe K et al. Intracellular temperature mapping with a fluorescent polymericthermometer and fluorescent lifetime imaging microscopy. Nat Commun 3, 705 (2012).
Virtual Reality Showcase for STELLARIS Confocal Microscopy Platform
Take a seat directly next to our experts in front of the microscope, configure various multicolor…Feb 21, 2023Read article
Confocal Imaging of Immune Cells in Tissue Samples
In this webinar, you will discover how to perform 10-color acquisition using a confocal microscope.…Dec 05, 2022Read article
Live-Cell Fluorescence Lifetime Multiplexing Using Organic Fluorophores
On-demand video: Imaging more subcellular targets by using fluorescence lifetime multiplexing…Nov 18, 2022Read article