X2Y scanner design
The tree-mirror concept is known as the “x2y-scanner”. This approach solves the general problem of scanner systems with two mirrors which cannot be placed in the correct position (the back focal plane) at the same time without additional relay optics. Usually, deviations are accepted by compromising the homogeneity of illumination, with negative effects on uniformity of brightness and resolution. A single mirror solution (gimbal/cardanic mounts) offers homogeneous illumination, but is restricted to very low scanning speeds.
The x2y scanner illuminates the full field of view homogeneously and allows high scanning speed. This scanning concept also allows the x-mirror to be exchanged by moving the alternative mirror (either conventional or resonant) into the proper position.
- Be careful not confuse it with the spinning-disc-type tandem scanner developed in the 1960s by Petran et al.: M. PETRÁŇ; M. HADRAVSKÝ; M.D. EGGER; R. GALAMBOS (1968), "Tandem-Scanning Reflected-Light Microscope", Journal of the Optical Society of America (JOSA), 58 (5): 661–664, DOI: 10.1364/JOSA.58.000661.
Virtual Reality Showcase for STELLARIS Confocal Microscopy Platform
Take a seat directly next to our experts in front of the microscope, configure various multicolor…Feb 21, 2023Read article
Confocal Imaging of Immune Cells in Tissue Samples
In this webinar, you will discover how to perform 10-color acquisition using a confocal microscope.…Dec 05, 2022Read article
Live-Cell Fluorescence Lifetime Multiplexing Using Organic Fluorophores
On-demand video: Imaging more subcellular targets by using fluorescence lifetime multiplexing…Nov 18, 2022Read article
Our confocal microscopes for top-class biomedical research provide imaging precision for subcellular…Visit related page